|Micro Tech Lab Professional solutions for digital micro and macro photography|
|An overview of the LM microscope adapters|
|We offer LM digital adapter solutions for the following microscopes|
|Online configurator: LM microscope adapter for all digital cameras and microscopes|
|LM Macroscope 24x (15x and 11x) for Focus Stacking: Highest resolution without compromise|
|LM Macroscope 9x (5x and 3x) for Focus Stacking: Highest resolution without compromise|
|LM photo microscopes: the flexible photography solution for large sensor cameras!|
|Special mounting medium for microscopy, non toxic,water solved,light hardening,fast solidifying and drying,neutraldoes not make air bubbles,high optical solution,color protecting,solvent free,high refractive index|
|Micro Tech Lab advisory service: microscopy, digital cameras, high quality photo|
|Microscope Digital Cameras: Camera ranking for microscopy use|
Are normal stereo microscopes suitable for high-quality photographic documentation?
95% of all stereo microscopes are designed either according to the Greenough principle or the telescope principle (Abbe principle). They deliver good to very good results on visual inspection. If, however, digital camera equipment is mounted onto a conventional stereo microscope, more often than not the quality of the images is disappointing. Especially in the case of the Greenough principle, the difference between the visual perception of the image quality and the image itself is particularly pronounced. How does such a difference arise? Is the human eye superior to the camera?
To answer this question we need to go into more detail.
The Greenough principle:
Greenough-type stereo microscopes have two completely separate optical paths. They strike the object with a difference in angle of 15º, which is created by two lenses situated in a common mount. The human brain is able to generate a three-dimensional image through the optical path and the angular offset. If you close one eye and look again through the microscope, you can see that only the centre of the image is optimally focussed. To the left and right of the centre, the image is either under- or overfocussed – in other words, there is a marked fuzziness. This effect is especially evident with planar objects, such as histological sections, geological stone sections or flat electronic components. The larger the optical path angle, the stronger the described effect. Optical systems have only a limited depth of field as a result of their physical characteristics. With Greenough stereo microscopes, structures are only depicted with optimum focus over a narrow area. This area shrinks with increasing magnification of the microscope. As the rays always hit the object at a fixed angle of 15º, there is always a sharpness gradient over the entire image field.
The image quality of the photographic documentation of planar objects can be improved by using a simple trick: The angular offset can be balanced out by tilting the object plane by 7.5°. This should cause the sample to be positioned at right angles to the optical path.
The telescope principle:
“Telescope style” (Abbe principle) stereo microscopes are more flexible than the Greenough-type microscopes described above. With this design there is no double lens; instead, the microscope has only one lens with a large diameter through which the optical paths run for both the left and the right eye. There is no real three-dimensional impression upon visual inspection through the common main lens, as there is with the Greenough instruments. This is particularly disadvantageous when working under the microscope, e.g. with specimens. The flaw in this system lies in the course of the optical path into the lens. As only one common lens is used, the lens edge must be used to generate the image. The intermediate image is at an infinite distance. The central, high-quality area of the lens unfortunately remains unused. However, due to its physical characteristics, the central part of the lens exhibits fewer optical errors than its edge. In photographic documentation, only a moderate image quality is achieved; nevertheless, this kind of microscope is better suited to microphotography than the Greenough microscope.
Conclusion: Stereo microscopes designed according to the more expensive telescope principle are indeed better suited than Greenough microscopes for high-level photographic documentation with film or digital cameras, but they, too, do not deliver optimum results.
All major manufacturers offer specific solutions for micro and macro photography. These instruments, however, tend to fall into the higher price bracket, starting from € 10,000.
The LM macroscopes with parallax-free, centric optical path and high-quality precision optics are replacing all conventional stereomicroscopes with parallel, non-centric optical paths in photographic documentation.
Convert your digital SLR camera into a professional microscope
Nikon D850 / Sony Alpha 7III / Sony Alpha 9 / Nikon D5 / Sony Alpha 7S II / Sony Alpha 7R III / Sony Alpha 7R II / Sony Alpha 7S / Canon EOS 5D Mark IV / Nikon D750 / Canon EOS 6D Mark II / Sony Alpha 6300 / Sony Alpha 6500 / Nikon D500 / Canon EOS 1D X / Nikon D810 / Nikon D4s / Nikon D800 / Nikon D800E / Nikon D4 / Nikon Df / Nikon D610 / Nikon D600 / Canon EOS 6D / Sony Alpha 99 II (SLT-A99 II) / Canon EOS 5DS R ( without low-pass filter) / Olympus OM-D E-M1 Mark II / Canon EOS 80D / Canon EOS 5DS / Sony Alpha 77 II / Canon EOS 70D / Nikon D7200 / Pentax K-1 Mark II / Canon EOS 200D / Canon EOS 800D / Rebel T7i / Canon EOS 77D / Canon EOS 5D Mark III / Canon EOS 60D / Sony Alpha 7R / Sony Alpha 7 / Nikon DS-Qi2 (Microscope Camera) /
|Modern DSLR on Olympus CH30/ CH40 microscope with LM microscope adapter|
|The Canon Camera Connect app in microscopy: wireless communication between smartphone or tablet and Canon EOS cameras|
|Test: Using the Canon EOS M50 on a microscope. Our LM microscope adapter solution allows for easy connection to either phototube or eyepiece tube|
|Microscope cameras on the Motic BA 210 and BA 310|
|Microscope cameras on the Motic SMZ 168 TP stereo zoom microscope|
|Controlling Sony Alpha cameras remotely in Live View mode from a tablet or smartphone: Sony’s PlayMemories Mobile app|
|The Alpha 6500 is another top-end system camera from Sony that performs impressively on the microscope|
|A comparison of CCD and CMOS sensors|
|Switching cameras made easy with the LM microscope adapter – just change the camera connection|
|The Pentax K-5 IIs DSLR camera tested on a microscope|
|Practical Test: Nikon D800|
|PC remote control of the Nikon Coolpix digital camera in microscopy|
|Controlling Olympus digital SLR cameras from your PC with the OLYMPUS Studio 2 software|
|An overview of the LM microscope adapters|
|Improved picture quality through mirror lock-up in SLR cameras|
|Using HD / UHD / 4k / 6k components in microphotography|
|Improving the quality of your microscope images|
|Customised adapter solutions: microscope adapters, photo microscopes, photo macroscopes|
|LM photo microscope for C-mount camera|
|LM photo microscope with Nikon C-mount camera DS-Fi2 and control panel DS-L3 for measurement tasks, long-term studies and lengthy observations|
|LM adapter solution for older Zeiss standard microscopes without photo tube|
|Preparing Zeiss SteREO Discovery microscopes (.V8, .V12,.V20) for microphotography|
|First-class microscopy images with Nikon Eclipse 55i / 50i and LM digital adapters|
|Which digital camera works best on a microscope?|
|Convert your digital SLR camera into a professional microscope|
|Special mounting media for microscopy|
Tips and tricks to connect your digital camera and to process digital images