Photographing through a microscope eyepiece
simplest case, it is possible to take freehand images through a
on the type of camera and lens used, with some samples, useable
results may be achieved.
for the hobbyist, if only a low quality photograph needs to be
produced, then this method is worth a try
1: Camera is held by hand over the eyepiece
it is evident that compact cameras and camcorders with lenses with
smaller front elements that are not telescopic are more suitable. The
best results are provided by very small compact cameras whose lenses
are integrated in the camera body.
camera lenses cause the most problems.
lenses mainly have larger maximum apertures and therefore have large
diameter lens elements.
results in peripheral darkening and some aberrations.
Due to the large number of lens elements in the lens unit the image
quality is considerably reduced.
problem with freehand photography via a microscope is posed by motion
when the hand is very steady, it will still move a few millimetres
here and there.
pressing the shutter release button in particular, the camera is
likely to shake.
using the shutter delay or self timer function on the camera, the
shutter release is automatically delayed for a few seconds.
trick reduces blurring.
right position for the digital camera over the microscope’s optical
axis is also not all that easy and can be quite arduous.
it can take several minutes to find the optimum position.
of taking freehand photographs over the eyepiece:
there is no optical adjustment between the microscope and the digital
unergonomic search for the correct camera position over the
image distortions No
optical adaptation to the size of the digital camera’s sensor
If superior image quality and ease of use is
important to you, then using an LM eyepiece adapter
is the best solution.
The best image quality is achieved by using a digital SLR camera with
Live View function without a lens unit. Only
the camera body is required.
The LM digital adapter projects the image directly onto the camera’s
sensor chip. No unnecessary
lens elements will then be in the light path.
The optimum LM digital
adapter is used according to the size of the camera’s sensor.
Caution! This manual is protected by copyright. Unauthorised reproduction thereof, in whole or in part, is punishable by law. Only use with MICRO TECH LABís approval.New LM Digital Adapter for:
Canon EOS 1D X / Nikon D4s / Nikon D4 / Canon EOS 6D / Canon EOS 70D / Nikon Df / Nikon D810 / Canon EOS 60D / Nikon D800 / Nikon D800E / Nikon D600 / Nikon D610 / Canon EOS 5D Mark III / Nikon D5300 (with automatic exposure chip ) / Canon EOS 1D Mark IV / Canon EOS 600D / Rebel T3i / Canon EOS 60Da for astrophotography / Canon EOS 650D / Rebel T4i / Canon EOS 700D / Rebel T5i / Nikon D3S / Canon EOS 5D Mark II / Canon EOS 7D / Canon EOS 550D / Rebel T2i / Kiss X4 Digital / Canon EOS 100D / Canon EOS 50D / Canon EOS 1200D / EOS Rebel T5 / EOS Kiss X70 / Nikon D3x / Canon EOS 1100D / Rebel T3 / Nikon D7000 / Nikon D7100 / Sony Alpha 7 / Sony Alpha 7R / Sony Alpha 7S / Olympus E-5 / Pentax K-5 / Canon EOS 40D / Pentax K-r / Sony SLT-A99 / Nikon D5200 / Nikon D5300 / Sony Alpha 6000 /
© 1999-2014 MICRO TECH LAB. All rights reserved